Install The installation package for the driver is distributed as an MSI 2. Now COM3 is associated with the hpexa driver session, which is associated with the logical name. Now if I run it, it does change the voltage level as I set but I keep getting either ‘RS framing error’, ‘Invalid character’ or ‘Invalid separator’. This is not how I prefer to do things but you can easily modify the initialize function to configure the com port there. Why am I getting this error?
|Date Added:||25 December 2013|
|File Size:||59.8 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 4 of 8. Using the Agilent test program I am able to set the output voltage.
agilent power supply control – Discussion Forums – National Instruments
Message 7 of 9. What exactly is stopping us from doing this? Also, if it’s a USB serial port, sometimes they get reassigned. Message 2 of 2. Message agjlent of 8. NET, Service Pack 2 is required.
Message 1 of 9. The logical name defines the ASL in your user application.
I added COM3 as a hardware asset for that driver session. Norbert Norbert CEO: If you can send it some commands with hyperterm, e36444a might give you a clue.
You can download the parser and its associated SDK from: Message 6 of 9. If you see an error about wscript. Agilenh Active Software Boards: Message 7 of 8.
What happens if you run an auto-zero and then take the measurement? Any insight or tips for resolving these issues would be appreciated as I keep searching for them. The VI that i wrote simply sets up the serial port, then in a loop queries the devices, reads the responses, and then writes the data to a file.
You can install the driver by double-clicking on the file. You can download the help lavbiew from: Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Message 8 of 9.
Agilent E36xx IVI-COM Driver
Message 5 of 9. I am very interesting in reading the KnowledgeBase article mentioned above, however, I receive a message saying that I do not have permission to read it. Do you have a quick guess?
In all probablility, you don’t need to modify any code. Message 1 of 6. I’m getting that error message, too. Most Active Hardware Boards: More Information For more information about this driver and other instrument drivers and software available from Agilent Technologies visit: Is laview content shown in the display similar?
Message 4 of 9.